On February 1, 2019, Stryker launched a voluntary field action on specific units of the LIFEPAK 15 Monitor/Defibrillators. The company notified LIFEPAK 15 customers of an issue that may cause the device to lock up after a defibrillation shock is delivered. The lock-up condition is defined as a blank monitor display with LED lights on, indicating power to the device, but no response in the keypad and device functions. A device in this condition has the potential to delay delivery of therapy. This delay in therapy has the potential to result in serious injury or death.
Stryker anticipated that all devices subject to this field action will be serviced by December 31, 2019.
Since the initial commercialization of LIFEPAK 15 in 2009, Stryker has become aware of 58 complaints reported globally for this issue, including six events in which the patient died following a delay in therapy. In all six of these cases, at least one shock was delivered prior to the device experiencing the lock-up condition. There are 13,003 devices potentially impacted by this issue and within scope of this field action.
The company instructed customers to continue to use their LIFEPAK 15 Monitor/Defibrillator according to the Operating Instructions until the correction can be completed. Device Automatic Self-Tests do not identify this fault, as it occurs during defibrillation. Customers should continue to perform the daily check as described in the Operator’s Checklist, specifically, the QUIK-COMBO therapy cable check as described in the General Maintenance and Testing Section (pages 10-4 and the LIFEPAK 15 Monitor/Defibrillator Operator’s Checklist, number 7). If a device exhibits the lockup condition during patient use, the steps from the General Troubleshooting Section (page 10-18) of the LIFEPAK 15 Monitor/Defibrillator Operating Instructions should be immediately followed.
See the FDA Recall
Medical Risk Law: Emergency Medical Services: Liability and Immunity for Medical Rescue
Medical Risk Law: When Urgency Leads to Errors: Liability for Emergency Care
Medical Risk Law: Diagnosis and Treatment of Heart Attacks: Liability Issues
Medical Risk Law: Unnecessary Cardiac Procedures: Getting to the Heart of the Risks
Medical Risk Law: Mending a Broken Heart: Malpractice Risks in Diagnosing and Treating Heart Disease
Medical Risk Law: Congenital Heart Conditions: How Infants, Adults, and Healthcare Providers Handle the Risks